|Date Added:||28 February 2004|
|File Size:||25.23 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Comparing the tick count to you wait seems dangerous to me. How do I sweep in agileent directions. Install Agilent Series Instrument Driver NI develops and supports thousands of instrument drivers for third party instruments to help engineers and scientists automate measurements.
Changing output amplitude on Agilent function generator while output is on Solved! Thanks last modified by shaobing Add an input to change the number of bursts you want the instrument to perform by right clicking the burst count terminal on the labvisw side of the Configure Burst node and selecting Change to control.
If the waveform is large and you are using RS interface, binary data format is not supported.
Customize the panel of the VI to fit your application needs. Back to Top 5. I opened the Configure Standard Waveform block and this is what it looks like: To avoid this, I decreased the output range of the function generator so as not to activate the switch.
Waveform Generators | Keysight Community
Is there any possibility that the waveform generator can only output a single period and than paused. I am trying to write a program to sweep amplitudes of a sine wave output from a function generator Agilent a. The 33xxx driver takes ms to change, and has a strange burst signal in between.
When performing frequency measurements of frequenciesi tie the time bases of both the standards and the counters together dependi Conduct Sequencing and Database Reporting With TestStand Test Management Software TestStand, a ready-to-run test management software, is designed to help you develop, execute, and deploy automated test and validation systems faster.
Using the sample code from the drivers, I modified them to generate a sine wave at 1Vpp, wait 1s, then change the amplitude to 0. Why is the output voltage twice the value I set?
Agilent 33XXX Series Signal Generator
I am using the Keysight B series waveform generator to frequency sweep a pulse waveform. NI PXI Waveform Generators can produce precise waveforms including sine, square, triangle, and ramp as well as arbitrary, user-defined waveforms by using data sequences or by streaming continuously from a host or peer-to-peer instrument within the PXI system.
Message 5 of 7. Open the Learning tab by selecting the Learn to Program tile in the Welcome screen or selecting the Learning tab from the top right.
LabVIEW driver for the A | Keysight (formerly Agilent’s Electronic Measurement)
Message 2 of 7. Thanks for the help answering my questions. NI develops and supports thousands of instrument drivers for third party instruments to help engineers and scientists automate measurements.
Keysight Technologies / Agilent Technologies 33210A
Windows is a registered trademark of Microsoft Corporation in the United States and other countries. From the Configuration Pane on the right, scroll down to the installed drivers section to find and install the NI Drop all the indicators and controls from the Unplaced items box on the panel. Message labvieww of 7.